Probe Electronics E2010
The electronic unit E2010 is the basic device for the use of ZIROX oxygen probes.
The signals of ZIROX probes are recorded with the E2010, processed to the oxygen concentration according to the NERNST equation and provided by standard interface (4-20 mA). The E2010 realizes the power supply and the heating control for heated ZIROX probes.
- Signal processing and measuring value display for ZIROX probes
- Measuring value output via standard interface
- Power supply and heating control for heated ZIROX probes
Technical Data Probe Electronics E2000
|Signal input||cell und thermal voltage ZIROX probes|
|Range||0...20.64 Vol.-% O2 (up to 100 Vol.-% on request)|
|Accuracy||rel. error < 5 %|
|Reference gas supply||none, must be supplied by user|
|Plain text display||LCD dot-matix, 2 x 16 characters|
|Keypad||2 membrane keys|
|Output||4...20 mA (corresponds to 0...20 Vol.-% O2)|
|Power supply||24 V DC +/- 10 %, 5 A|
|Dimensions||96 x 96 x 125 mm3|
|Weight||approx. 650 g|
|Protection degree||IP30 (Front IP52)|
|Working temperature||0...+40 °C, 0...95 % rel. humidity|
|Storage temperature||-20...+50 °C, 0...95 % rel. humidity|
Note: You are not sure which device is required for your application?
Please complete our questionnaire and send it to infoziroxde.